MOQ : 1 Piece
|Country of Origin||India|
|Stage Time Sequences||12|
|No. of S.S. Basket Rotor||2|
|S.S. Tissue Capsules||24|
|S.S. Beakers without lip||10|
|S.S. Wax Bath||2|
|Capacity||1 ltr / 2 ltr|
|Working Voltage||220 V|
Preferred Buyer From
|Location||India, Bangladesh, Nepal, Sri Lanka|
Under the guidance of eminent pathologists our R & D Division has developed yet another unique model of Tissue Processor having complete Programmer logic Controller for complete cycle. Timing & stage sequence, Wax bath temperature safety system etc. are all controlled/operated by feather touch key board with Digital Display. The entire unit looks very attractive with aesthetic appeal. The data is retained in case of power failure. Front panel comprises of the key board, digital display, control switch. The PID temperature controller have been used for controlling the wax bath temperature.
Technical Data: The 12 stage timing sequence may be programmed for each stage for the duration between 1 minute to 9 hrs. 59 minute, in steps of 1 minute.
Delay Time: Delay time can be programmed upto 99 hours and will be sequenced with main control.
Wax Bath Safety Control: Wax Bath PID temperature controller gives signal if the temperature is not between 50° to 65°C. Safety System will function at 10th, 11th & 12th position and will stop the main unit in stage 9 onward till the set temperature is attained in the wax bath.
Basket Rotor: Slowly rotating 2 basket rotors are also operated through the microprocessor.
The unit is supplied complete with 2 Nos. S.S. Basket Rotors, 2 Nos. S.S. Baskets, 24 Nos. S.S. Tissue Capsules, 10 Nos. S.S. beakers without lip, 2 Nos. S.S. Wax Baths (Special designed for use with PID temperature controller). Suitable to work on 220 V, single phase, 50 Hz, AC supply. Cap. : 1 litre, 2 liters